Differentiate, distinguishes, compare and explain what the main difference between standpipe, standpipe piezometer and piezometer. Comparison and Differences.
A standpipe normally contains plastic pipes with perforated holes at the base. The annular space between the perforated tube and casing is filled with gravel or sand backfill. Under such an arrangement, standpipe is used to measure water level of a certain region.
A standpipe piezometer is a type of piezometer which measures pore water pressure at a certain level. It consists of plastic pipes without holes. The tip of the standpipe piezometer is perforated and the annular space between the tip of the piezometer and soil is filled with sand while the annular space between other parts of plastic tube and soil is filled with cement/bentonite grout to seal off water from entering the region of piezometer tip. This enables the pore water pressure in the region of piezometer tip to be measured. In essence, standpipe piezometers are installed to study the pore water pressure of a specified depth below ground. However, it suffers from the disadvantage that the response time is relatively slow in clayey soils.
Search More Topics on the Web
Difference between piezometer vs standpipe, standpipe piezometer
stvspipe, stvspipe piezometer vs piezometer
Differences between piezometer vs standpipe, standpipe piezometer